You are in our Global Site
1. What is ultrasoundUltrasound is not much different from the sound waves we are familiar with, except that we can't hear its "sound". When the frequency of sound waves reaches more tha...Read More >
Many customers think that the cleaning solution is just water before purchasing an ultrasonic cleaner. When they know that different cleaning fluids are used, some people still have no idea about the ...Read More >
1. Wireless handheld ultrasound deviceIn surgery, "Handheld Wireless Ultrasound" is not limited by the cable and is more flexible to use. Moreover, the probe can be put into a disposable pro...Read More >
The diasonograph generates incident ultrasound waves (transmitted waves) and receives reflected ultrasound waves (echoes) through probes. It is an important part of diagnostic equipment. The high-frequency electric energy excites the crystal in the probe to produce mechanical vibration, and the mechanical vibration of the reflected ultrasound wave can be converted into an electric pulse by the probe. That is to say, the probe can convert electric energy into sound energy, and it can also convert sound energy into electric energy, so the probe is also called an ultrasound transducer. The principle comes from the piezoelectric effect of the crystal.
Crystals with piezoelectric effect properties are called piezoelectric crystals. At present, the crystal sheets commonly used in portable usg probe include artificial or natural crystals such as lead zirconate, barium titanate, quartz, and lithium sulfate. Barium titanate and lead zirconate are polycrystalline ceramic bodies sintered at high temperatures. After the blank is sintered into a ceramic body, it is properly ground and trimmed to obtain the required geometric dimensions. After polarization by a high-voltage direct current electric field, it has piezoelectric properties and becomes ultrasound transducer parts.
Positive piezoelectric effect
In a certain direction of the crystal or ceramic, mechanical force is applied to make it deform, and the two force-bearing surfaces of the crystal or ceramic produce charges with opposite signs; the direction of deformation is opposite, the polarity of the charge changes accordingly, and the charge density is proportional to the applied mechanical force. This effect of stimulating the surface charge due to the mechanical force is called the positive piezoelectric effect.
Inverse piezoelectric effect
Applying a voltage along the direction of the electric field on the surface of the crystal or ceramic will cause the geometric shape of the crystal or ceramic to be strained under the action of the electric field. The direction of the voltage changes and the direction of the strain also changes. The deformation is proportional to the electric field voltage. The deformation effect induced by the electric field is called the inverse piezoelectric effect.
In general, the piezoelectric effect is linear. However, when the electric field is too strong or the pressure is high, a nonlinear relationship will appear.
Crystals and ceramics have different natural frequencies of mechanical vibration due to the different cutting positions and geometric dimensions. When the frequency of the applied alternating voltage is consistent with the natural frequency, the mechanical vibration generated is the strongest; when the frequency of the applied mechanical force is consistent with the natural frequency, the generated charge is also the most. The frequency of the excitation pulse in the ultrasound diagnostic apparatus must be the same as the natural frequency of the ultrasound transducer.
Call us on:+86-028-67878343-5